|Alarm after 20 seconds no movement|
|No cords, wires, pads, or external power|
|Clips onto nappy / diaper|
Why Snuza® Pico?
• Measure Abdominal Breathing Movement, Body Temperature, Body Position
• Fall alert, if the baby takes a tumble
• Small – clips onto baby’s diaper
• Portable – cordless baby monitor
• Streams information to a smartphone or tablet
Frequently Asked Questions:
Can I connect two smart devices to my Snuza® Pico at the same time? No. Only one smart device can be connected to your Snuza® Pico at the same time.
I cannot register my Snuza® Pico. Please check if your spouse or partner have not already registered it. Or if you have a second hand monitor* please contact the previous owner of the Snuza® Pico and request that they deactivate the Snuza® Pico. Log in with previous users details at connect.snuza.com/devices and click on the red de-activate button at the bottom of the page and then register the monitor in your name. *We do not promote the purchase of second hand devices due to hygiene and quality reasons.
Why is the temperature not accurate when I first put the Snuza® Pico on? When you first put the Snuza® Pico on, it is slightly colder than your baby and tends to reduce the temperature of the skin around the Snuza® Pico. Leave the Snuza® Pico on the baby for 10 – 15 minutes in order to limit this effect.
Why is the Snuza® Pico not staying connected to the phone and what can I do about it? Bluetooth Low Energy technology has limited range but you can do some simple things to improve this:
Why am I getting false breathing alarms? The Snuza® Pico is designed to sense abdominal movement and so if there is no movement detected it will alarm. To avoid false alarms ensure you have attached Snuza® Pico firmly to the nappy as indicated in the Quick Start Guide and place the Snuza® Pico where the baby will not lie on the device.
The Snuza® Pico alarms while I’m feeding or changing the baby! There is a pause function on the Snuza® Pico that can be activated from the smart phone or by un-clipping the Snuza® Pico and briefly pressing the power button.
Can the app be loaded onto multiple phones to simultaneously pair with one Snuza® Pico? The app can be loaded onto multiple phones but only one phone can be connected to the Snuza® Pico at a time.
Can multiple Snuza® Pico’s be monitored from the same phone? Yes, up to a maximum of three Snuza® Pico’s can be connected to one smart phone or tablet.
I didn’t receive the registration email? Please first check your junk mail folder if you don’t find the email try login with your username, it will offer to send the confirmation email again. Alternately you could go to http://connect.snuza.com/send-confirmation and enter the email you used to register. If the email is not found simply register with it. If you still experience problems please send your email address and username to email@example.com so we can investigate.
How do I connect a second user to the Snuza® Pico? The easiest way is to log into the second phones Snuza app with the same username and password (note only one phone can be connected to the Snuza® Pico at a time). Alternately, the Snuza® Pico can be shared with a second registered user via the Snuza Connect website connect.snuza.com by clicking on “Devices” and sending an invite by entering an email address under “Secondary Users”. This invite will need to be accepted by the recipient, and the recipient must have an account.
The Snuza® Pico won’t connect, what can I do? Check that the Snuza® Pico is on. Check if the blue connection light is flashing, if it is, check that other smart devices in the house are not connected to it. Try exiting or swiping the app away and restarting it. Try resetting the Snuza® Pico by holding the power button for longer than 10 seconds, remember to turn it on again. If all of these don’t help send your serial number to firstname.lastname@example.org so we can investigate.
The graphs have disappeared from the iPhone app how do I get them back? With the Snuza® Pico connected to the app, click on the baby’s name or picture to navigate to the settings page. Then switch on the appropriate switches under display settings. Alternatively, you can go to the bottom of the settings page and click the defaults button.
The Snuza® Pico is falling off the diaper. First make sure that the top lever is seated correctly, then ensure that the thin flap of the diaper is folded over the ticker part, which is what the clip grips on to.
The Clip is very loose and not seated properly. The clip is designed to be replaceable and will unseat if moved more than 60 degrees past the closed position. There is a friction stop at 60 degrees, but with a little force, it can be moved beyond this. To re-seat the clip. please follow this How To video:
The ‘tick’ functionality has gone. The firmware is continuously being improved to make the product better. As part of this process, a new breathing detection algorithm was developed which is vastly more effective than before. The downside is that the ‘tick’ functionality has temporarily been removed while we work on making it compatible with the new improved algorithm.
Read the basic tips for using your Snuza® Pico smart wearable movement monitor below.
Snuza® Pico is a mobile movement monitor so the non-contact sensor must always be in contact with the baby’s tummy. The non-contact sensor however will still detect movement if it loses contact with the tummy for a few seconds.
To prevent unnecessary false alarms caused by Snuza® Pico being out of range of abdominal movement:
Please note: Snuza® abdominal movement monitors detect the slightest movement, including movement from external sources such as when travelling in a vehicle, pushing a pram, etc. Snuza® Pico should only be relied upon when there is no external movement present, i.e. when the vehicle, pram, carry cot, etc. is stationery.
Tips for attaching the Snuza® Pico
Attaching the Snuza® Pico to clothing
If you follow these guidelines properly, you can rest assured that baby’s every movement is being monitored.
Recharging the Battery
The Snuza® Pico makes use of a rechargeable battery and is a sealed unit.
To charge the Snuza® Pico, place it in the USB charging station (Supplied with the Snuza® Pico) for 2 hours.
A full charge allows for up to 40 hours of usage.
|Nett:||Approximately 40g (unit)|
|Gross:||Approximately 200g (packaged)|
|Battery:||(rechargeable) Lithium Polymer (LiPo)|
|Noise level of alarm:||85 dBm at 10cm|
|Materials used:||Medical grade polycarbonate FDA approved thermoplastic-elastomer.|
|Certification:||CE (2001/95/EC, 89/336/EEC) (TV Rheinland)|
|RoHS (Restriction of Hazardous Substances Directive - Europe)|
|FCC (Federal Communications Commission - USA)|
|ISO9001:2008 - manufactured in an ISO9001:2008 certified facility|
|Important Snuza® Pico Tips — Download|
|Quick Start Guide — Download|
|Snuza® Pico Manual — Download|
|Snuza® Pico APP Terms & Conditions — Download|